Technical Data ISOTEST 6
From 20 to 500Vdc infinitely variable, is generated in the device!
From 0 to 200 MΩ and ∞; with every test voltage.
9V battery (function from 5 to 10V) current: approx. 3-60mA.
No zero adjustable by stabilised test voltage necessarily.
130 x 70 x 55 mm (5,1 x 2,8 x 2,2 inch)
with battery: 330 g (11,6 oz)
By the infinitely variable adjustable test voltage the
6" is suitable very well for capacitor check and isolation check in
electric devices and arrangements. Before every introduction after
a battery test should be carried out: Voltage knob in position "500
V", Rx clips short-circuit, a tracer press, pointer must reach
"0", otherwise a battery change is due. Draw off in addition lower
case bowl, battery exchange, cases close, battery test. Voltage changes
low-spirited a tracer are absolutely to be avoided, the announcement
could be thereby damaged.
Danger of the test voltage:
Electric shocks are to be feared in no voltage- area by
the low test
current of max. 0.05 mA with 500 V (source resistance > 10
in condenser check:
By the loading of the capacitor disagreeable and dangerous electric
appear from 60 V, please around special care!
Applications of the "ISOTEST 6":
Capacitors (no Electrolytics!) variable condensors, trimmers, diodes,
tubes (intern leakance), glow discharge devices (neon lamps),
thyristors, to engine, electric devices, installations, cables etc. -
in each case allowed test voltage!
Ideally by capacitors would always be announcement
Coupling capacitors on radios and amplifier should be always
flawless capacitors (resistance ∞ endlessly).
With capacitor measurements always wait for shutdown of
The greater the capacity, the longer the measuring time.